timers
twin stim comb tens ems digital unit with timer 3162
twin stim tens and ems combo our most popular digital electrotherapy unit the twin stim gives the user the choice of 5 modes of tens and 2 modes of ems twin stims features include dual channel 7 modes timer patient compliance meter 100ma output safety amplitude cover sturdy metal belt clip rubber side railings for maximum handling channels dual channel isolated between channels pulse amplitude adjustable 0 100 ma peak into 500 ohm load each channel voltage 0 50v load 500 ohm pulse rate adjustable from 2 to 150 hz 1 hz step pulse width adjustable from 50 to 300 microseconds 10us step wave form asymmetrical bi phasic square pulse on time adjustable 2 & x7e ; 90 seconds 1 sec step off time adjustable 2 & x7e ; 90 seconds 1 sec step ramp time adjustable 1 & x7e ; 8 seconds 1 sec step the on time will increase and decrease in the setting value 3 ems modes constant synchronous alternate constant stimulation based on setting value only pulse width pulse rate and timer are adjustable in this mode constant is equal to the normal mode of a tens unit synchronous stimulation of both channels occurs synchronously 5 modes b burst n normal m modulation sd1 strength duration sd2 system includes stimulator lead wires 1 package of high quality self adhesive electrodes 9v battery hard plastic carrying case & instruction booklet sku comsu594
complete medical supplies