food thermometers
fluke fluke 53 2 b thermocouple thermometer 1 input
thermocouple thermometer thermocouple type type k j t e r s n number of inputs 1 probe connector mini type temp . range f type j 346 degrees to 2192 degrees f type k 328 degrees to 2501.6 degrees f type t 418 degrees to 752 degrees f type e 238 degrees to 1832 degrees f type n 328 degrees to 2372 degrees f type r and type s 32 degrees to 3212.6 degrees f br b features b ul li accuracy 1.1 degrees c 2 degrees f li li battery type 3 aa li li features electronic offset function allows compensation of thermocouple errors to maximize overall accuracy readout in degree c degree f or kelvin k splash and dust resistant case protected by impact absorbing holster user friendly front panel is easy to set up and operate sleep mode increases battery life typical 1000 hr . battery life battery door allows easy battery replacement without breaking the calibration seal optional toolpak accessory allows the thermometer to hang from any metal object recall function allows logged data to be easily reviewed on the meter display ir communication port allows data to be exported to optional flukeview forms r temperature pc software sold separately for further analysis and graphing li li includes 1 mfr . no . fluke 80pk 1 bead probe thermocouple li li item thermocouple thermometer li li temp . range f type j 346 degrees to 2192 degrees f type k 328 degrees to 2501.6 degrees f type t 418 degrees to 752 degrees f type e 238 degrees to 1832 degrees f type n 328 degrees to 2372 degrees f type r and type s 32 degrees to 3212.6 degrees f li li data logging 500 points li li display backlit 3 line lcd li li probe connector mini type li li resolution 0.1 degrees c li li sampling rate 1 sec . 10 sec . 1 min . 10 min . li li temp . range c type j 210 degrees c to 1200 degrees c type k 200 degrees c to 1372 degrees c type t 250 degrees c to 400 degrees c type e 150 degrees c to 1000 degrees c type n 200 degrees c to 1300 degrees c type r and type s 0 degrees c to 1767 degrees c li li manufacturers warranty length 3 yr . li li data hold yes li li auto power off yes li li number of inputs 1 li li memory storage 500 li li mode log hold min max avg li li memory min max avg dif li li thermocouple type type k j t e r s n li li probe included 1 li li differential t1 t2 no li li thermocouple type 2 j k t e n r s li ul
fluke