cable ties
ysi do200acc 04 dissolved oxygen meter4m cable g7244596
dissolved oxygen meter dissolved oxygen range 0 to 20 ppm oxygen concentration 0 to 200 percent waterproof ip67 automatic temperature compensation no displayed temperature range 6 degrees to 46 degrees c 21.2 degrees to 114.8 degrees f salinity range 0 to 40 ppt operating temp . range f 21.2 degrees to 114.8 degrees operating temp . range c 6 degrees to 46 degrees display digital auto power off yes memory 50 data sets temperature compensation range 6 degrees to 46 degrees c br b features b ul li includes probe carrying case 4m cable battery li li item dissolved oxygen meter li li operating temp . range f 21.2 degrees to 114.8 degrees li li display digital li li power source 9v battery li li range 0 to 20 ppm li li temp . accuracy 0.3 degrees c 1 digit li li temp . resolution 0.1 degrees c li li manufacturers warranty length 12 mo. li li waterproof ip67 li li auto power off yes li li calibration yes li li memory 50 data sets li li operating temp . range c 6 degrees to 46 degrees li li interface rs 232 li li temperature compensation range 6 degrees to 46 degrees c li li dissolved oxygen range 0 to 20 ppm li li dissolved oxygen range accuracy 2 % of the reading or 0.2 ppm whichever is greater li li oxygen concentration 0 to 200 % li li oxygen concentration accuracy 2 % of the reading or 2 % air saturation whichever is greater li li salinity range 0 to 40 ppt li li dissolved oxygen resolution 0.01 ppm li li oxygen concentration resolution 0.1 % air saturation li li salinity accuracy 1 % of the reading or 0.1 ppt whichever is greater li li testing parameter dissolved oxygen temperature li li cable length 4m li li automatic temperature compensation no li li displayed temperature range 6 degrees to 46 degrees c 21.2 degrees to 114.8 degrees f li ul