showerheads & handheld showers
gems sensors ft 330 0.2 2.0 gpm flow rate sensor 3 8 mnpt 0.2 to 2.0 gpm
flow rate sensor turbine housing material noryl connection size 3 8 mnpt in . max . voltage 24 pulses per gallon 10313 flow range 0.2 to 2.0 gpm accuracy plus or minus 2 % cable length 3 ft. input power 5 to 24 vdc 8ma flow range 0.8 to 7.6 lpm max . viscosity 32 81 ssu operating pressure 200 psi operating temp . range 4 to 176 f circuit type npn repeatability 0.5 % wetted materials nylon ss noryl pom ps agency compliance nsf std . 61 listed includes cable br b features b ul li cable length ft. 3 li li circuit type npn li li connection size in . 3 8 mnpt li li flow range 0.2 to 2.0 li li gauge conductor 22 awg li li height in . 57 64 li li includes cable li li item flow rate sensor li li length in . 2 15 64 li li max . voltage 24 li li operating pressure psi 200 li li operating temp . range f 4 to 176 li li type turbine li li width in . 45 64 li li flow range gpm 0.2 to 2.0 li li housing material noryl li li pulses per gallon 10313 li li wetted materials nylon ss noryl pom ps li li agency compliance nsf std . 61 listed li li output frequency hertz 34 to 343 li li accuracy % 2 li li max . viscosity ssu 32 to 81 li li burst pressure psi 1000 li li recommended filtration microns less than 50 li li input power 5 to 24 vdc 8ma li li flow range lpm 0.8 to 7.6 li li pulses per liter 2724 li li repeatability % 0.5 li ul