pipe benders
dwyer instruments 628 10 ch p1 e5 s1 pressure transmitter 0 100 psi conduit
pressure transmitter pressure range 0 100 psi accuracy 1 % full scale aluminum conduit housing output 4 20 ma power required 13 to 30 vdc connection 1 4 in mnpt compatible with liquids and gas material of construction 316 ss and 316l ss operating temp 0 to 200 f nema rating 4x loop resistance 0 to 1300 ohms max pressure 200 psi thermal error 0.04 % full scale per degree f includes instructions br b features b ul li accuracy 1 % full scale li li electrical connections terminal board with 1 2 npt female conduit connection li li features li li for use with li li housing cast aluminum li li includes instructions li li item pressure transducer li li operating temp . 0 to 200 f li li pressure range in . wc li li display li li display units li li max . pressure psi 200 li li nema rating 4x li li output 4 to 20 ma dc li li range 0 to 100 psi li li safety rating li li manufacturers warranty length 1 year li li wetted materials 316 ss and 316l ss li li lead length in . li li process connection 1 4 mnpt li li calibration certificate li li compatible with liquids and gas li li loop resistance 0 to 1300 ohms li li power required 13 to 30 vdc li li thermal error 0.02 % full scale per degree f li ul