cable ties
ysi 20 4 meter cable4m cable g6500067
meter cable electric conductivity range 0 to 200 ms cm auto range tds range 0 to 100g l tds constant range 0.30 to 1.00 0.65 default dissolved oxygen range 0 to 50mg l oxygen concentration 0 to 500 percent air saturation waterproof ip67 automatic temperature compensation no displayed temperature range 5 degrees to 55 degrees c eg11 0 degrees to 45 degrees c do compensation range for mg l salinity range 0 to 70 ppt operating temp . range f 21.2 degrees to 114.8 degrees br b features b ul li includes 4m cable li li item meter cable li li operating temp . range f 21.2 degrees to 114.8 degrees li li display no li li power source power cable li li range 0 to 50mg l li li temp . accuracy 0.2 degrees c li li temp . resolution 0.1 degrees c li li manufacturers warranty length 24 mo. li li waterproof ip67 li li auto power off no li li calibration yes li li memory 50 data sets li li operating temp . range c 6 degrees to 46 degrees li li interface military spec connector li li temperature compensation range 0 degrees to 45 degrees c li li dissolved oxygen range 0 to 50mg l li li dissolved oxygen range accuracy 0 to 20mg l 2 % of the reading or 0.2 mg l whichever is greater 20 to 50mg l 6 % of the reading li li oxygen concentration 0 to 500 % air saturation li li oxygen concentration accuracy 0 to 200 % air saturation 2 % of the reading or 2 % air saturation whichever is greater ; 200 to 500 % air saturation 6 % of the reading li li tds range 0 to 100g l tds constant range 0.30 to 1.00 0.65 default li li tds resolution 0.001 0.01 or 0.1 g l li li tds accuracy dependent on temperature and conductivity calculated from those parameters li li electric conductivity range 0 to 200 ms cm auto range li li electric conductivity resolution 0.0001 to 0.1 ms cm range dependent li li electric conductivity accuracy 1 or 4 m cable 1.0 % of the reading or 1.0 us cm whichever is greater 10 20 or 30 m cable 2.0 % of the reading or 1.0 us cm whichever is greater 1 or 4 m cable 1.0 % of the reading or 1.0 us cm whichever is greater 10 20 or 30 m cable 2.0 % of the reading or 1.0 us cm whichever is greater li li salinity range 0 to 70 ppt li li dissolved oxygen resolution 0.01 or 0.1 mg l li li oxygen concentration resolution 0.1 % or 1 % air saturation user selectable li li salinity resolution 0.01 ppt li li salinity accuracy 1 % of the reading or 0.1 ppt whichever is greater li li barometric pressure range 500 to 800 mmhg li li testing parameter dissolved oxygen temperature li li cable length 4m li li automatic temperature compensation no li li displayed temperature range 5 degrees to 55 degrees c eg11 0 degrees to 45 degrees c do compensation range for mg l li ul