pressure gauges
dwyer instruments 628 08 gh p1 e1 s1 transmitter 0 30 psi 36 in lead
pressure transmitter pressure range 0 30 psi accuracy 1 % full scale stainless steel housing output 4 20 ma power required 13 to 30 vdc connection 1 4 in mnpt compatible with liquids and gas material of construction 316 ss and 316l ss operating temp 0 to 200 f nema rating 4x lead length 36 in loop resistance 0 to 1300 ohms max pressure 60 psi thermal error 0.04 % full scale per degree f includes instructions br b features b ul li thermal error 0.02 % full scale per degree f li li operating temp . 0 to 200 f li li loop resistance 0 to 1300 ohms li li includes instructions li li accuracy 1 % full scale li li power required 13 to 30 vdc li li item pressure transducer li li max . pressure psi 60 li li nema rating 4x li li lead length in . 36 li li output 4 to 20 ma dc li li housing stainless steel li li compatible with liquids and gas li li process connection 1 4 mnpt li li electrical connections 36 wire li li range 0 to 30 psi li li manufacturers warranty length 1 year li li wetted materials 316 ss and 316l ss li li features li li for use with li li pressure range in . wc li li display li li display units li li safety rating li li calibration certificate li ul