door handle sets
ashcroft k17mo242c15000 xfm pressure transducerrange 0 to 5000 psi
pressure transducer range 0 to 5000 psi accuracy 0.5 % full scale output 4 to 20 ma at 12 to 30vdc power required 9 to 36vdc process connection 1 4 in npt male electrical connections 1 2 in conduit lead length 36 in wetted materials 316 stainless steel housing stainless steel operating temp 20 to 180 f nema rating 4x non incendive safety rating class 1 div . 2 groups a b c and d max pressure 7500 psi includes instructions br b features b ul li accuracy 0.5 % full scale li li electrical connections 1 2 conduit li li features li li for use with li li housing stainless steel li li includes instructions li li item pressure transducer li li operating temp . 20 to 180 f li li pressure range in . wc li li display li li display units li li max . pressure psi 7500 li li nema rating 4x non incendive li li output 4 to 20 ma at 12 to 30vdc li li range 0 to 5000 psi li li safety rating class 1 div . 2 groups a b c and d li li manufacturers warranty length 1 year li li wetted materials 316 stainless steel li li lead length in . 36 li li process connection 1 4 npt male li li calibration certificate li li compatible with any process compatible with 316stainless steel li li loop resistance dependent on loop supply voltage li li power required 9 to 36vdc li li thermal error .04 % per degree f from 20 to 168f li ul