cable ties
ysi pro2030 handheld conductivity do meter no cable
handheld conductivity do meter electric conductivity range 0 to 200 ms cm tds range 0 to 100 g l dissolved oxygen range 0 to 50 mg l oxygen concentration 0 to 500 % salinity range 0 to 70 ppt temp . range f 23 degree to 131 degree display graphic backlit lcd barometric pressure range 400 to 999.9 mmhg cable length order cable separately power source 2 alkaline c batteries includes batteries and instructions manufacturers warranty length 3 years br b features b ul li temp . resolution 0.1 degrees c li li item handheld conductivity do meter li li temp . range c 5 degrees to 55 degrees li li manufacturers warranty length 3 years li li auto power off yes li li temperature compensation range 0 degrees to 45 degrees c li li includes batteries and instructions li li memory 50 data sets li li temp . accuracy 0.3 degrees c li li dissolved oxygen range 0 to 50 mg l li li dissolved oxygen range accuracy 2 % of the reading or 0.2 mg l whichever is greater up to 20 mg l 6 % of reading from 20 to 50 mg l li li oxygen concentration 0 to 500 % li li oxygen concentration accuracy 2 % of reading or 2 % air saturation whichever is greater up to 200 % 6 % of reading from 200 to 500 % li li tds range 0 to 100 g l li li tds resolution 0.0001 0.01 or 0.1 g l li li tds accuracy dependent on conductivity and temperature li li electric conductivity range 0 to 200 ms cm li li electric conductivity resolution 0.0001 to 0.1 ms cm range dependent li li electric conductivity accuracy 0.5 % of reading or 0.001 ms cm whichever is greater li li display graphic backlit lcd li li power source 2 alkaline c batteries li li temp . range f 23 degrees to 131 degrees li li dissolved oxygen resolution 0.01 or 0.1 mg l user selectable li li oxygen concentration resolution 0.1 % or 1 % air saturation user selectable li li salinity resolution 0.01 ppt li li salinity accuracy 1 % of reading or 0.1 ppt whichever is greater li li salinity range 0 to 70 ppt li li cable length order cable separately li li barometric pressure range 400 to 999.9 mmhg li li operating temp . range f 23 degrees to 131 degrees li li waterproof yes li li operating temp . range c 5 degrees to 55 degrees li li sub category handheld li li automatic temperature compensation yes li li displayed temperature range 5 to 55 degrees c li ul