pressure regulators
flomec om006a511 222g5 flowmeter oval gear 220 psi 1 4 in .
flowmeter oval gear housing material aluminum fitting size in . 1 4 flow material oil chemicals high viscosity thick fluids lubricants flow range gph 0.5 to 27 flow range gpm 0.008 to 0.45 flow range lpm 0.033 to 1.67 fitting type fnpt accuracy percent 1 wetted materials aluminum 316 stainless steel carbon viton pressure rating psi 220 fluid temp . range deg . f 5 to 140 max . viscosity 1000 cps length in . 2.7 frequency range hz 0.53 to 29.81 br b features b ul li fitting size in . 1 4 li li fitting type fnpt li li flow range 0.5 to 27 gph 2 to 100 lph li li includes local display pulse output li li item flowmeter li li length in . 2.7 li li pressure rating psi 220 li li type oval gear li li display type lcd li li display units gallon liter li li flow range gpm 0.008 to 0.45 li li frequency range hz 0.53 to 29.81 li li housing material aluminum li li repeatability 0.03 % of reading li li sensor type reed switch and hall effect li li special features totals cumulative and batch rate of flow pulse output battery power external power li li wetted materials aluminum 316 stainless steel carbon viton li li max . viscosity 1000 cps li li output type npn open collector pulse li li strainer 200 mesh recommended li li accuracy % 1 li li o ring viton li li rotor type stainless steel oval gears li li power requirement 9v battery included li li fluid temp . range deg . c 20 to 60 li li flow range 5 cps 2.7 to 27 gph 10 to 100 lph li li ambient temperature 20 degrees to 60 degrees c li li typical k factor reed switch 3975 pulses per gallon hall effect 3975 pulses per gallon li li fluid temp . range deg . f 5 to 140 li li flow range lpm 0.033 to 1.67 li li flow material oil chemicals high viscosity thick fluids lubricants li li flow range gph 0.5 to 27 li ul